The Synergy HT Multi-detection reader is a robotic-compatible microplate .. and J. Sambrook () Molecular Cloning A Laboratory Manual, Cold Spring. View and Download BioTek Synergy HT operator’s manual online. Multi-Mode Microplate Reader. Synergy HT Laboratory Equipment pdf manual download. Manuals and User Guides for BioTek Synergy HT. We have 2 BioTek Synergy HT manuals available for free PDF download: Operator’s Manual.
Page Clean the Internal Components Stand facing the front of the instrument. Enter text from picture: Contact BioTek Customer Care with any questions.
The knurled plastic should sit flush against the hanger surface, as shown below. Page 40 Page 41 Page 42 Page 43 – 8: Run The Absorbance Plate Test Remove the Shipping Panel Page 35 Page 36 – 3: Cleaning The Reader’s Internal Surface Microplate carrier, fully extended Moisten do not soak a clean cotton cloth with alcohol, water, or with water and mild detergent.
Do not allow water or other cleaning solution to run into the interior of the instrument. Page Clean the Internal Components Lift the ground wire and move it off to the side. The risk factors associated with your assays may require that the Operational and Performance Qualification procedures be performed more frequently than shown above.
Remove two screws from the top center of the rear panel Remove two side screws When reinstalling the shroud, press down firmly on the top to maintain a good light seal while tightening the top screws. The installed syringes should resemble the following: Chapters Table Of Contents Only qualified technical personnel should perform troubleshooting and service procedures on internal components. Repeat steps 4—9 using the Synergy HT Dispenser 2. Page 54 Perform these steps to prepare the dispense module for shipment: The illustration should facilitate system setup with a robotic instrument, such as the BioStack User Safety Part 1: The trough holds up to 1.
Page 40 Lift out the module and place it on a level surface. Every Synergy HT reader and external dispense module is ut tested at BioTek prior to shipment and they should operate properly upon initial setup.
Plug the lamp cables into the power source located to the right of the lamp. After reassembling the instrument, perform the following to verify that the instrument is functioning properly: Set the shroud aside.
Repackaging And Shipping Applies only to Synergy HT models with injectors. See page 91 for instructions for analyzing the results. Clean both sides of the filter with lens paper. Biitek slowest of the three available modes.
Do not apply lubricants to the microplate carrier or carrier track. Connect the Host Computer Page 48 – Installing The Time-resolved Fluorescence Cartridge Results Kanual If any tests fail, prime the fluid lines and rerun the test s. Page 37 – 4: Creating Protocols And Experiments Removing The Reader’s Shroud Page 84 Delta 1: Place the tip priming trough in the left-rear pocket of the carrier.
Test Solution Recipes Page Appendix E: When reinstalling the internal dispense tubes, be sure to align the tubing ports with the injector heads as shown in this diagram.